12/19/23 16:35:51.537	Information	Image Studio Version 5.2.5	GMT -5:00
12/19/23 16:35:51.537	Information	OS: Windows 8 6.2	GMT -5:00
12/19/23 16:35:51.537	Information	Java Home: c:\program files\licor\image studio\jre	GMT -5:00
12/19/23 16:35:51.553	Information	JRE Version: 1.7.0_72 32-bit	GMT -5:00
12/19/23 16:35:51.553	Information	Java library: C:\Program Files\Licor\Image Studio;C:\WINDOWS\Sun\Java\bin;C:\WINDOWS\system32;C:\WINDOWS;C:\Program Files (x86)\Common Files\Oracle\Java\javapath;C:\WINDOWS\system32;C:\WINDOWS;C:\WINDOWS\System32\Wbem;C:\WINDOWS\System32\WindowsPowerShell\v1.0\;C:\WINDOWS\System32\OpenSSH\;C:\Program Files\PolicyPak\;C:\Users\wmao2\AppData\Local\Microsoft\WindowsApps;;c:\program files\licor\image studio\jre\bin;. 32-bit	GMT -5:00
12/19/23 16:35:51.600	Information	Database migrations: 	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Analysis Key: Grid_Analysis	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Analysis Key: Image_Studio_Base	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Analysis Key: Shapes	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Analysis Key: Western_Analysis	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Instrument Key: CDG_Acquire	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Instrument Key: Odyssey_Acquire	GMT -5:00
12/19/23 16:35:51.616	Information	Installed Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/19/23 16:35:51.631	Information	Installed Instrument Key: OdysseyFc_Acquire	GMT -5:00
12/19/23 16:35:51.631	Information	Installed Instrument Key: OdysseySa_Acquire	GMT -5:00
12/19/23 16:35:51.631	Information	Installed Instrument Key: Pearl_Acquire	GMT -5:00
12/19/23 16:35:51.631	Information	Selected Instrument Key: OdysseyCLx_Acquire	GMT -5:00
12/19/23 16:35:51.631	Information	Loading project C:\Users\wmao2\2023-12-19\Images\...	GMT -5:00
12/19/23 16:35:52.022	Information	Loading more acquisitions...	GMT -5:00
12/19/23 16:35:52.037	Information	USBCat Client Version 1.0.2	GMT -5:00
12/19/23 16:35:52.459	Information	Have initialized and added 5 Acquisitions to Work Area...	GMT -5:00
12/19/23 16:35:52.459	Information	Finished adding Acquisitions to Work Area.	GMT -5:00
12/19/23 16:35:52.522	Information	Done updating tables.	GMT -5:00
12/19/23 16:35:52.522	Information	Image Studio Version 5.2.5	GMT -5:00
12/19/23 16:35:56.068	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 16:35:56.068	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 16:35:56.130	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 16:35:56.161	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 16:35:56.161	Information	[Instrument Request]  1 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 6.0 700 800 	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 6 700 800	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Y mapping: [y0=0,y1=6] => [top=34,bottom=2874] h=355	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 y_scan_pixels=355, passes=2 => lines_to_scan=710	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Blocks=25	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=255	GMT -5:00
12/19/23 16:35:59.145	Information	[Instrument Response] 1 Progress 0% 6 minutes	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 1 Result 0 OK	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    659 mV 	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    723 mV 	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               25 C  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:35:59.161	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/19/23 16:35:59.176	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 16:35:59.176	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 16:35:59.176	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 16:35:59.176	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/19/23 16:35:59.176	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.0 V  	GMT -5:00
12/19/23 16:35:59.223	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 16:36:04.269	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/19/23 16:36:21.515	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 16:36:29.576	Information	[Instrument Request]  2 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 4.9 700 800 	GMT -5:00
12/19/23 16:36:29.591	Information	[Instrument Response] 2 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 4.9 700 800	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 Blocks=20	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=209	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:36:29.623	Information	[Instrument Response] 2 Result 0 OK	GMT -5:00
12/19/23 16:36:30.716	Information	Starting acquisition...	GMT -5:00
12/19/23 16:36:31.653	Information	[Instrument Request]  3 ScanAuto 169um medium 0.0 0.1 5.0 0.0 4.9 700 800 	GMT -5:00
12/19/23 16:36:31.763	Information	[Instrument Response] 3 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 4.9 700 800	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 Blocks=20	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=209	GMT -5:00
12/19/23 16:36:31.778	Information	[Instrument Response] 3 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:36:31.794	Information	[Instrument Response] 3 Progress 0% Initializing	GMT -5:00
12/19/23 16:36:31.794	Information	[Instrument Response] 3 Status 0 Preparation complete	GMT -5:00
12/19/23 16:36:34.684	Information	[Instrument Response] 3 Status 0 Motors initialized	GMT -5:00
12/19/23 16:36:34.731	Information	[Instrument Response] 3 Status 0 X-Motor configured	GMT -5:00
12/19/23 16:36:34.731	Information	[Instrument Response] 3 Status 0 DSP configured	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 Cooling	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 290 xofs=1 yofs=0	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 290 xofs=0 yofs=0	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 start=34 scan_lines=15 line=0 end=290 700=Low,L1 800=High,4	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 start=34 scan_lines=30 line=0 end=290 700=High,4 800=Low,L1	GMT -5:00
12/19/23 16:36:34.778	Information	[Instrument Response] 3 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 16:36:35.293	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 16:36:35.340	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:36:38.293	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:36:39.402	Information	[Instrument Response] 3 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:36:40.308	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:36:40.355	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:36:40.355	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:36:40.355	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:36:40.355	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:36:40.355	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:36:45.322	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/19/23 16:36:47.337	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:36:47.384	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         129 V  	GMT -5:00
12/19/23 16:36:47.384	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:36:47.384	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:36:47.384	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:36:48.337	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/19/23 16:36:48.400	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:36:49.337	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:36:51.352	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:36:52.352	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:36:54.367	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:36:54.711	Information	[Instrument Response] 3 ImageBlock 700 0 0 290 15 290 290 8700	GMT -5:00
12/19/23 16:36:54.711	Information	[Instrument Response] 3 ImageBlock 800 0 0 290 15 290 290 8700	GMT -5:00
12/19/23 16:36:54.726	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:36:54.758	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:36:55.367	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:37:00.444	Information	[Instrument Response] 3 Progress 8% 4 minutes	GMT -5:00
12/19/23 16:37:01.397	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/19/23 16:37:01.444	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:37:01.444	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:37:01.444	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:37:02.396	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:37:08.083	Information	[Instrument Response] 3 ImageBlock 700 0 15 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:08.083	Information	[Instrument Response] 3 ImageBlock 800 0 15 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:08.083	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:37:08.114	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:37:13.441	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      315 mA 	GMT -5:00
12/19/23 16:37:14.441	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:37:14.503	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/19/23 16:37:14.503	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:37:14.503	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:37:15.456	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:37:21.439	Information	[Instrument Response] 3 ImageBlock 700 0 30 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:21.439	Information	[Instrument Response] 3 ImageBlock 800 0 30 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:21.455	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:37:21.486	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:37:21.548	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:37:21.548	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:37:22.470	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:37:22.533	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:37:24.485	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:37:25.485	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:37:27.500	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         123 V  	GMT -5:00
12/19/23 16:37:27.563	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:37:27.563	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:37:27.563	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          57 mA 	GMT -5:00
12/19/23 16:37:28.500	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:37:29.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/19/23 16:37:31.062	Information	[Instrument Response] 3 Progress 19% 3 minutes, 30 seconds	GMT -5:00
12/19/23 16:37:34.811	Information	[Instrument Response] 3 ImageBlock 700 0 45 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:34.811	Information	[Instrument Response] 3 ImageBlock 800 0 45 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:34.811	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:37:34.842	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:37:40.560	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:37:40.607	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         151 V  	GMT -5:00
12/19/23 16:37:40.607	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:37:40.607	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:37:40.607	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:37:41.559	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/19/23 16:37:41.606	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:37:42.559	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:37:48.167	Information	[Instrument Response] 3 ImageBlock 700 0 60 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:48.167	Information	[Instrument Response] 3 ImageBlock 800 0 60 290 15 290 290 8700	GMT -5:00
12/19/23 16:37:48.183	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:37:48.214	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:37:48.589	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:37:48.636	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:37:49.589	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:37:49.636	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:37:51.588	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:37:52.604	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:37:54.603	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/19/23 16:37:54.666	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:37:54.666	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:37:54.666	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:37:55.619	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:38:01.196	Information	[Instrument Response] 3 Progress 31% 3 minutes	GMT -5:00
12/19/23 16:38:01.539	Information	[Instrument Response] 3 ImageBlock 700 0 75 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:01.539	Information	[Instrument Response] 3 ImageBlock 800 0 75 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:01.539	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:38:01.571	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:38:07.663	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:38:07.725	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/19/23 16:38:07.725	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:38:07.725	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:38:08.678	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:38:14.896	Information	[Instrument Response] 3 ImageBlock 700 0 90 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:14.911	Information	[Instrument Response] 3 ImageBlock 800 0 90 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:14.911	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:38:14.943	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:38:18.707	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:38:19.723	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:38:20.722	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         144 V  	GMT -5:00
12/19/23 16:38:20.769	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 16:38:20.769	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:38:20.769	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:38:20.769	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:38:21.722	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/19/23 16:38:21.785	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:38:21.785	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:38:22.722	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:38:28.268	Information	[Instrument Response] 3 ImageBlock 700 0 105 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:28.268	Information	[Instrument Response] 3 ImageBlock 800 0 105 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:28.268	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:38:28.299	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:38:31.345	Information	[Instrument Response] 3 Progress 43% 2 minutes, 30 seconds	GMT -5:00
12/19/23 16:38:34.782	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:38:34.844	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/19/23 16:38:34.844	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:38:34.844	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:38:35.797	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:38:41.640	Information	[Instrument Response] 3 ImageBlock 700 0 120 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:41.640	Information	[Instrument Response] 3 ImageBlock 800 0 120 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:41.640	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:38:41.655	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:38:45.826	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:38:46.842	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:38:47.841	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/19/23 16:38:47.888	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:38:47.888	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:38:47.888	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:38:48.841	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:38:54.996	Information	[Instrument Response] 3 ImageBlock 700 0 135 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:54.996	Information	[Instrument Response] 3 ImageBlock 800 0 135 290 15 290 290 8700	GMT -5:00
12/19/23 16:38:54.996	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:38:55.027	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:39:00.729	Information	[Instrument Response] 3 Progress 54% 2 minutes	GMT -5:00
12/19/23 16:39:00.901	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 16:39:00.948	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         136 V  	GMT -5:00
12/19/23 16:39:00.948	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:39:00.948	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:39:00.948	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:39:01.901	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/19/23 16:39:01.963	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:39:02.900	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:39:08.368	Information	[Instrument Response] 3 ImageBlock 700 0 150 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:08.368	Information	[Instrument Response] 3 ImageBlock 800 0 150 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:08.368	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:39:08.399	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:39:11.430	Information	[Instrument Response] 3 Progress 58% 1 minute, 50 seconds	GMT -5:00
12/19/23 16:39:12.945	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:39:13.945	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:39:14.960	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:39:15.007	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:39:15.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:39:15.007	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:39:20.975	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:39:21.396	Information	[Instrument Response] 3 Progress 62% 1 minute, 40 seconds	GMT -5:00
12/19/23 16:39:21.724	Information	[Instrument Response] 3 ImageBlock 700 0 165 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:21.724	Information	[Instrument Response] 3 ImageBlock 800 0 165 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:21.724	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:39:21.756	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:39:21.974	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:39:28.020	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:39:28.067	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         110 V  	GMT -5:00
12/19/23 16:39:28.067	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:39:28.067	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:39:29.020	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:39:31.347	Information	[Instrument Response] 3 Progress 66% 1 minute, 30 seconds	GMT -5:00
12/19/23 16:39:35.096	Information	[Instrument Response] 3 ImageBlock 700 0 180 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:35.096	Information	[Instrument Response] 3 ImageBlock 800 0 180 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:35.096	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:39:35.128	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:39:40.814	Information	[Instrument Response] 3 Progress 70% 1 minute, 20 seconds	GMT -5:00
12/19/23 16:39:41.064	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         129 V  	GMT -5:00
12/19/23 16:39:41.126	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:39:41.126	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:39:41.126	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:39:42.079	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 16:39:43.079	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:39:48.103	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:39:48.463	Information	[Instrument Response] 3 ImageBlock 700 0 195 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:48.463	Information	[Instrument Response] 3 ImageBlock 800 0 195 290 15 290 290 8700	GMT -5:00
12/19/23 16:39:48.463	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:39:48.494	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:39:49.103	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:39:51.149	Information	[Instrument Response] 3 Progress 74% 1 minute, 10 seconds	GMT -5:00
12/19/23 16:39:55.133	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:39:55.195	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 16:39:55.195	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:39:55.195	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:39:56.133	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:40:01.491	Information	[Instrument Response] 3 Progress 77% 1 minute	GMT -5:00
12/19/23 16:40:01.819	Information	[Instrument Response] 3 ImageBlock 700 0 210 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:01.819	Information	[Instrument Response] 3 ImageBlock 800 0 210 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:01.819	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:40:01.850	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:40:08.192	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         105 V  	GMT -5:00
12/19/23 16:40:08.239	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:40:08.239	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:40:08.239	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:40:09.192	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:40:11.442	Information	[Instrument Response] 3 Progress 81% 50 seconds	GMT -5:00
12/19/23 16:40:15.191	Information	[Instrument Response] 3 ImageBlock 700 0 225 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:15.191	Information	[Instrument Response] 3 ImageBlock 800 0 225 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:15.191	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:40:15.222	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:40:15.269	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:40:15.269	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:40:16.222	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:40:16.269	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:40:20.908	Information	[Instrument Response] 3 Progress 85% 40 seconds	GMT -5:00
12/19/23 16:40:21.236	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:40:21.299	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         124 V  	GMT -5:00
12/19/23 16:40:21.299	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:40:21.299	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:40:22.252	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:40:23.252	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:40:28.547	Information	[Instrument Response] 3 ImageBlock 700 0 240 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:28.547	Information	[Instrument Response] 3 ImageBlock 800 0 240 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:28.563	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:40:28.578	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:40:31.250	Information	[Instrument Response] 3 Progress 89% 30 seconds	GMT -5:00
12/19/23 16:40:34.296	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         150 V  	GMT -5:00
12/19/23 16:40:34.358	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         159 V  	GMT -5:00
12/19/23 16:40:34.358	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:40:34.358	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:40:34.358	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:40:35.296	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/19/23 16:40:35.358	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:40:35.358	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:40:36.311	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:40:41.201	Information	[Instrument Response] 3 Progress 93% 20 seconds	GMT -5:00
12/19/23 16:40:41.919	Information	[Instrument Response] 3 ImageBlock 700 0 255 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:41.919	Information	[Instrument Response] 3 ImageBlock 800 0 255 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:41.919	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:40:41.950	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:40:42.325	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:40:42.388	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:40:43.325	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:40:43.388	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:40:45.340	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:40:46.340	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:40:48.355	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:40:48.402	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         103 V  	GMT -5:00
12/19/23 16:40:48.402	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:40:48.402	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:40:49.355	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:40:51.152	Information	[Instrument Response] 3 Progress 96% 10 seconds	GMT -5:00
12/19/23 16:40:55.276	Information	[Instrument Response] 3 ImageBlock 700 0 270 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:55.276	Information	[Instrument Response] 3 ImageBlock 800 0 270 290 15 290 290 8700	GMT -5:00
12/19/23 16:40:55.291	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:40:55.307	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:40:57.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         136 V  	GMT -5:00
12/19/23 16:40:57.447	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:40:57.447	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:40:57.447	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:40:58.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 16:40:59.400	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:41:01.102	Information	[Instrument Response] 3 Progress 100% 0 seconds	GMT -5:00
12/19/23 16:41:01.415	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:41:01.462	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         126 V  	GMT -5:00
12/19/23 16:41:01.462	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:41:01.462	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:41:02.415	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:41:03.102	Information	[Instrument Response] 3 ImageBlock 700 0 285 290 5 290 290 2900	GMT -5:00
12/19/23 16:41:03.102	Information	[Instrument Response] 3 ImageBlock 800 0 285 290 5 290 290 2900	GMT -5:00
12/19/23 16:41:03.102	Information	[Instrument Response] 3 Status 0 COMPLETE SCAN 263.702 seconds	GMT -5:00
12/19/23 16:41:03.118	Information	Transferring image block at (0,285) with size 290x5 to the 700 channel.	GMT -5:00
12/19/23 16:41:03.118	Information	[Instrument Response] 3 Status 0 Averages (sec): setup=2.00005 line=0.378794 => total=222	GMT -5:00
12/19/23 16:41:03.133	Information	Transferring image block at (0,285) with size 290x5 to the 800 channel.	GMT -5:00
12/19/23 16:41:03.149	Information	[Instrument Response] 3 Status 0 Comparisons: setup=1.00997 line=0.939617	GMT -5:00
12/19/23 16:41:03.196	Information	[Instrument Response] 3 Status 0 Finalize motors	GMT -5:00
12/19/23 16:41:03.196	Information	[Instrument Response] 3 Progress 100% Scan complete	GMT -5:00
12/19/23 16:41:03.227	Information	[Instrument Response] 3 Result 0 OK	GMT -5:00
12/19/23 16:41:03.227	Information	Acquisition 0000032_01 finished with channels: 700 800.	GMT -5:00
12/19/23 16:41:03.539	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 16:41:03.539	Information	Scan completed successfully	GMT -5:00
12/19/23 16:41:03.680	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         129 V  	GMT -5:00
12/19/23 16:41:03.696	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          81 V  	GMT -5:00
12/19/23 16:41:03.711	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 16:41:03.711	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 16:41:03.711	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 16:41:04.430	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          77 V  	GMT -5:00
12/19/23 16:41:04.477	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          49 V  	GMT -5:00
12/19/23 16:41:05.439	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          46 V  	GMT -5:00
12/19/23 16:41:05.480	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          30 V  	GMT -5:00
12/19/23 16:41:06.431	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          30 V  	GMT -5:00
12/19/23 16:41:06.479	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          19 V  	GMT -5:00
12/19/23 16:41:07.439	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          20 V  	GMT -5:00
12/19/23 16:41:07.494	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          13 V  	GMT -5:00
12/19/23 16:41:08.450	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          13 V  	GMT -5:00
12/19/23 16:41:08.513	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           9 V  	GMT -5:00
12/19/23 16:41:09.437	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/19/23 16:41:09.500	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/19/23 16:41:10.453	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           7 V  	GMT -5:00
12/19/23 16:41:10.514	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/19/23 16:41:11.443	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           4 V  	GMT -5:00
12/19/23 16:41:11.505	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           2 V  	GMT -5:00
12/19/23 16:41:13.459	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/19/23 16:41:16.475	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:41:17.478	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:41:54.469	Information	[Instrument Request]  4 Stop 	GMT -5:00
12/19/23 16:41:54.516	Information	[Instrument Response] 4 Result 0 OK	GMT -5:00
12/19/23 16:41:55.516	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
12/19/23 16:42:31.305	Information	Have discovered 1 instruments.	GMT -5:00
12/19/23 16:42:31.305	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/19/23 16:42:31.305	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 16:42:31.305	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 16:42:31.383	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 16:42:31.414	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 16:42:31.430	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 16:42:31.430	Information	[Instrument Request]  5 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 4.9 700 800 	GMT -5:00
12/19/23 16:42:33.382	Information	Starting acquisition...	GMT -5:00
12/19/23 16:42:34.273	Information	[Instrument Request]  6 ScanAuto 169um medium 0.0 0.1 5.0 0.0 4.9 700 800 	GMT -5:00
12/19/23 16:42:34.382	Information	[Instrument Response] 5 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 4.9 700 800	GMT -5:00
12/19/23 16:42:34.382	Information	[Instrument Response] 5 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:42:34.382	Information	[Instrument Response] 5 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:42:34.382	Information	[Instrument Response] 5 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 Blocks=20	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=209	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 5 Result 0 OK	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 4.9 700 800	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 Blocks=20	GMT -5:00
12/19/23 16:42:34.398	Information	[Instrument Response] 6 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=209	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 6 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 6 Progress 0% Initializing	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 6 Status 0 Preparation complete	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    666 mV 	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/19/23 16:42:34.413	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.1 V  	GMT -5:00
12/19/23 16:42:34.445	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 16:42:37.397	Information	[Instrument Response] 6 Status 0 Motors initialized	GMT -5:00
12/19/23 16:42:37.460	Information	[Instrument Response] 6 Status 0 X-Motor configured	GMT -5:00
12/19/23 16:42:37.460	Information	[Instrument Response] 6 Status 0 DSP configured	GMT -5:00
12/19/23 16:42:37.460	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 Cooling	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 290 xofs=1 yofs=0	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 290 xofs=0 yofs=0	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 start=34 scan_lines=15 line=0 end=290 700=Low,L1 800=High,4	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 start=34 scan_lines=30 line=0 end=290 700=High,4 800=Low,L1	GMT -5:00
12/19/23 16:42:37.506	Information	[Instrument Response] 6 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 16:42:38.412	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:42:38.459	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:42:41.287	Information	[Instrument Response] 6 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:42:41.427	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         104 V  	GMT -5:00
12/19/23 16:42:41.474	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    717 mV 	GMT -5:00
12/19/23 16:42:41.474	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         162 V  	GMT -5:00
12/19/23 16:42:41.474	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:42:41.474	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:42:41.474	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:42:41.474	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:42:42.427	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:42:42.490	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 16:42:42.490	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:42:46.442	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/19/23 16:42:47.442	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:42:48.457	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:42:49.457	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:42:49.504	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         115 V  	GMT -5:00
12/19/23 16:42:49.504	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:42:49.504	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:42:50.457	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:42:52.472	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:42:52.519	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 16:42:53.472	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:42:56.611	Information	[Instrument Response] 6 ImageBlock 700 0 0 290 15 290 290 8700	GMT -5:00
12/19/23 16:42:56.611	Information	[Instrument Response] 6 ImageBlock 800 0 0 290 15 290 290 8700	GMT -5:00
12/19/23 16:42:56.611	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:42:56.643	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:42:59.501	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      315 mA 	GMT -5:00
12/19/23 16:43:02.329	Information	[Instrument Response] 6 Progress 8% 4 minutes	GMT -5:00
12/19/23 16:43:02.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         136 V  	GMT -5:00
12/19/23 16:43:02.563	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:43:02.563	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:43:02.563	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:43:02.563	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:43:03.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 16:43:03.579	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:43:04.516	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:43:04.578	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:43:09.983	Information	[Instrument Response] 6 ImageBlock 700 0 15 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:09.983	Information	[Instrument Response] 6 ImageBlock 800 0 15 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:09.983	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:43:10.015	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:43:16.576	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:43:16.623	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:43:16.623	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:43:16.623	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:43:19.575	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 16:43:22.590	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:43:23.340	Information	[Instrument Response] 6 ImageBlock 700 0 30 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:23.340	Information	[Instrument Response] 6 ImageBlock 800 0 30 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:23.340	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:43:23.371	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:43:23.590	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:43:23.652	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:43:29.635	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/19/23 16:43:29.682	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:43:29.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:43:29.682	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:43:30.635	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:43:33.338	Information	[Instrument Response] 6 Progress 19% 3 minutes, 30 seconds	GMT -5:00
12/19/23 16:43:36.712	Information	[Instrument Response] 6 ImageBlock 700 0 45 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:36.712	Information	[Instrument Response] 6 ImageBlock 800 0 45 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:36.712	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:43:36.743	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:43:42.679	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:43:42.742	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         130 V  	GMT -5:00
12/19/23 16:43:42.742	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:43:42.742	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:43:42.742	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:43:43.695	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          93 V  	GMT -5:00
12/19/23 16:43:43.741	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:43:44.694	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:43:46.694	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:43:47.709	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:43:49.709	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:43:49.771	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:43:50.068	Information	[Instrument Response] 6 ImageBlock 700 0 60 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:50.068	Information	[Instrument Response] 6 ImageBlock 800 0 60 290 15 290 290 8700	GMT -5:00
12/19/23 16:43:50.068	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:43:50.099	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:43:50.709	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:43:50.771	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:43:56.754	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/19/23 16:43:56.801	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:43:56.801	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:43:56.801	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:43:57.754	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:44:03.096	Information	[Instrument Response] 6 Progress 31% 3 minutes	GMT -5:00
12/19/23 16:44:03.440	Information	[Instrument Response] 6 ImageBlock 700 0 75 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:03.440	Information	[Instrument Response] 6 ImageBlock 800 0 75 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:03.440	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:44:03.471	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:44:09.798	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:44:09.861	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         106 V  	GMT -5:00
12/19/23 16:44:09.861	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:44:09.861	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:44:10.798	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:44:16.797	Information	[Instrument Response] 6 ImageBlock 700 0 90 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:16.797	Information	[Instrument Response] 6 ImageBlock 800 0 90 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:16.797	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:44:16.828	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:44:16.890	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 16:44:19.843	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:44:20.842	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:44:20.905	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:44:22.858	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         123 V  	GMT -5:00
12/19/23 16:44:22.904	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:44:22.904	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:44:22.904	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:44:23.857	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:44:25.873	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      285 mA 	GMT -5:00
12/19/23 16:44:30.168	Information	[Instrument Response] 6 ImageBlock 700 0 105 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:30.168	Information	[Instrument Response] 6 ImageBlock 800 0 105 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:30.168	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:44:30.200	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:44:33.230	Information	[Instrument Response] 6 Progress 43% 2 minutes, 30 seconds	GMT -5:00
12/19/23 16:44:35.902	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         156 V  	GMT -5:00
12/19/23 16:44:35.964	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         151 V  	GMT -5:00
12/19/23 16:44:35.964	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:44:35.964	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:44:35.964	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:44:36.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:44:36.964	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          88 V  	GMT -5:00
12/19/23 16:44:36.964	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:44:37.917	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:44:43.525	Information	[Instrument Response] 6 ImageBlock 700 0 120 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:43.525	Information	[Instrument Response] 6 ImageBlock 800 0 120 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:43.540	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:44:43.556	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:44:43.947	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:44:44.946	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:44:46.946	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:44:47.961	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:44:49.961	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         102 V  	GMT -5:00
12/19/23 16:44:50.023	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:44:50.023	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:44:50.023	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:44:50.976	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:44:56.897	Information	[Instrument Response] 6 ImageBlock 700 0 135 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:56.897	Information	[Instrument Response] 6 ImageBlock 800 0 135 290 15 290 290 8700	GMT -5:00
12/19/23 16:44:56.897	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:44:56.928	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:45:02.614	Information	[Instrument Response] 6 Progress 54% 2 minutes	GMT -5:00
12/19/23 16:45:03.020	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:45:03.083	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         119 V  	GMT -5:00
12/19/23 16:45:03.083	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:45:03.083	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:45:04.036	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:45:10.253	Information	[Instrument Response] 6 ImageBlock 700 0 150 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:10.269	Information	[Instrument Response] 6 ImageBlock 800 0 150 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:10.269	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:45:10.284	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:45:13.331	Information	[Instrument Response] 6 Progress 58% 1 minute, 50 seconds	GMT -5:00
12/19/23 16:45:14.065	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:45:15.080	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:45:16.080	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         144 V  	GMT -5:00
12/19/23 16:45:16.127	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         152 V  	GMT -5:00
12/19/23 16:45:16.127	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:45:16.127	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:45:16.127	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:45:17.080	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          85 V  	GMT -5:00
12/19/23 16:45:17.142	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:45:17.142	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:45:18.079	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:45:19.095	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:45:21.094	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:45:23.281	Information	[Instrument Response] 6 Progress 62% 1 minute, 40 seconds	GMT -5:00
12/19/23 16:45:23.625	Information	[Instrument Response] 6 ImageBlock 700 0 165 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:23.625	Information	[Instrument Response] 6 ImageBlock 800 0 165 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:23.625	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:45:23.656	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:45:30.139	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:45:30.202	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          98 V  	GMT -5:00
12/19/23 16:45:30.202	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:45:30.202	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:45:31.155	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:45:33.248	Information	[Instrument Response] 6 Progress 66% 1 minute, 30 seconds	GMT -5:00
12/19/23 16:45:36.997	Information	[Instrument Response] 6 ImageBlock 700 0 180 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:36.997	Information	[Instrument Response] 6 ImageBlock 800 0 180 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:36.997	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:45:37.013	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:45:42.715	Information	[Instrument Response] 6 Progress 70% 1 minute, 20 seconds	GMT -5:00
12/19/23 16:45:43.199	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         114 V  	GMT -5:00
12/19/23 16:45:43.246	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:45:43.246	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:45:43.246	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:45:44.199	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:45:46.214	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:45:47.214	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:45:50.353	Information	[Instrument Response] 6 ImageBlock 700 0 195 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:50.353	Information	[Instrument Response] 6 ImageBlock 800 0 195 290 15 290 290 8700	GMT -5:00
12/19/23 16:45:50.353	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:45:50.385	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:45:53.056	Information	[Instrument Response] 6 Progress 74% 1 minute, 10 seconds	GMT -5:00
12/19/23 16:45:56.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         152 V  	GMT -5:00
12/19/23 16:45:56.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         136 V  	GMT -5:00
12/19/23 16:45:56.305	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:45:56.305	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:45:56.305	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:45:57.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/19/23 16:45:57.321	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:45:58.258	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:46:03.382	Information	[Instrument Response] 6 Progress 77% 1 minute	GMT -5:00
12/19/23 16:46:03.725	Information	[Instrument Response] 6 ImageBlock 700 0 210 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:03.725	Information	[Instrument Response] 6 ImageBlock 800 0 210 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:03.725	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:46:03.741	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:46:10.318	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:46:10.365	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:46:10.365	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:46:10.365	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:46:13.317	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:46:13.379	Information	[Instrument Response] 6 Progress 81% 50 seconds	GMT -5:00
12/19/23 16:46:14.332	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:46:17.082	Information	[Instrument Response] 6 ImageBlock 700 0 225 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:17.082	Information	[Instrument Response] 6 ImageBlock 800 0 225 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:17.082	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:46:17.113	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:46:22.815	Information	[Instrument Response] 6 Progress 85% 40 seconds	GMT -5:00
12/19/23 16:46:23.377	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:46:23.440	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         109 V  	GMT -5:00
12/19/23 16:46:23.440	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:46:23.440	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:46:24.377	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:46:30.454	Information	[Instrument Response] 6 ImageBlock 700 0 240 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:30.454	Information	[Instrument Response] 6 ImageBlock 800 0 240 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:30.454	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:46:30.485	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:46:33.141	Information	[Instrument Response] 6 Progress 89% 30 seconds	GMT -5:00
12/19/23 16:46:36.421	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         129 V  	GMT -5:00
12/19/23 16:46:36.484	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:46:36.484	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:46:36.484	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:46:36.484	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:46:37.437	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          93 V  	GMT -5:00
12/19/23 16:46:37.483	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:46:38.436	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:46:43.091	Information	[Instrument Response] 6 Progress 93% 20 seconds	GMT -5:00
12/19/23 16:46:43.451	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:46:43.513	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:46:43.810	Information	[Instrument Response] 6 ImageBlock 700 0 255 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:43.810	Information	[Instrument Response] 6 ImageBlock 800 0 255 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:43.826	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:46:43.857	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:46:44.466	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:46:44.513	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:46:50.499	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:46:50.544	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          91 V  	GMT -5:00
12/19/23 16:46:50.545	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:46:50.545	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:46:51.503	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:46:53.062	Information	[Instrument Response] 6 Progress 96% 10 seconds	GMT -5:00
12/19/23 16:46:57.173	Information	[Instrument Response] 6 ImageBlock 700 0 270 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:57.173	Information	[Instrument Response] 6 ImageBlock 800 0 270 290 15 290 290 8700	GMT -5:00
12/19/23 16:46:57.188	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:46:57.204	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:46:59.532	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         118 V  	GMT -5:00
12/19/23 16:46:59.579	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:46:59.579	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:46:59.579	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:47:00.531	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:47:03.000	Information	[Instrument Response] 6 Progress 100% 0 seconds	GMT -5:00
12/19/23 16:47:03.546	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:47:03.593	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         111 V  	GMT -5:00
12/19/23 16:47:03.593	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:47:03.593	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:47:04.546	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:47:04.999	Information	[Instrument Response] 6 ImageBlock 700 0 285 290 5 290 290 2900	GMT -5:00
12/19/23 16:47:04.999	Information	[Instrument Response] 6 ImageBlock 800 0 285 290 5 290 290 2900	GMT -5:00
12/19/23 16:47:05.015	Information	Transferring image block at (0,285) with size 290x5 to the 700 channel.	GMT -5:00
12/19/23 16:47:05.015	Information	[Instrument Response] 6 Status 0 COMPLETE SCAN 263.714 seconds	GMT -5:00
12/19/23 16:47:05.030	Information	Transferring image block at (0,285) with size 290x5 to the 800 channel.	GMT -5:00
12/19/23 16:47:05.046	Information	[Instrument Response] 6 Status 0 Averages (sec): setup=2.00005 line=0.378814 => total=222	GMT -5:00
12/19/23 16:47:05.093	Information	[Instrument Response] 6 Status 0 Comparisons: setup=1.00997 line=0.939566	GMT -5:00
12/19/23 16:47:05.093	Information	[Instrument Response] 6 Status 0 Finalize motors	GMT -5:00
12/19/23 16:47:05.093	Information	[Instrument Response] 6 Progress 100% Scan complete	GMT -5:00
12/19/23 16:47:05.124	Information	[Instrument Response] 6 Result 0 OK	GMT -5:00
12/19/23 16:47:05.124	Information	Acquisition 0000033_01 finished with channels: 700 800.	GMT -5:00
12/19/23 16:47:05.483	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 16:47:05.483	Information	Scan completed successfully	GMT -5:00
12/19/23 16:47:05.593	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         113 V  	GMT -5:00
12/19/23 16:47:05.608	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          71 V  	GMT -5:00
12/19/23 16:47:05.608	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 16:47:05.608	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 16:47:05.608	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 16:47:06.562	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          68 V  	GMT -5:00
12/19/23 16:47:06.612	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          43 V  	GMT -5:00
12/19/23 16:47:07.562	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          42 V  	GMT -5:00
12/19/23 16:47:07.617	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          26 V  	GMT -5:00
12/19/23 16:47:08.567	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          28 V  	GMT -5:00
12/19/23 16:47:08.622	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          18 V  	GMT -5:00
12/19/23 16:47:09.577	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          19 V  	GMT -5:00
12/19/23 16:47:09.624	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          12 V  	GMT -5:00
12/19/23 16:47:10.564	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/19/23 16:47:10.626	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/19/23 16:47:11.582	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/19/23 16:47:11.629	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/19/23 16:47:12.585	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/19/23 16:47:12.632	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/19/23 16:47:14.606	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           2 V  	GMT -5:00
12/19/23 16:47:15.631	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/19/23 16:47:16.596	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      254 mA 	GMT -5:00
12/19/23 16:47:19.602	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:47:32.833	Information	[Instrument Response] 0 Lid Open	GMT -5:00
12/19/23 16:47:59.124	Information	[Instrument Request]  7 Stop 	GMT -5:00
12/19/23 16:47:59.171	Information	[Instrument Response] 7 Result 0 OK	GMT -5:00
12/19/23 16:47:59.827	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
12/19/23 16:48:20.900	Information	Have discovered 1 instruments.	GMT -5:00
12/19/23 16:48:20.900	Information	Discovered Name: CLX-2217 IP Address: fe80:0:0:0:201:29ff:fe8d:a2af%14	GMT -5:00
12/19/23 16:48:20.900	Information	Attempting to connect to instrument CLX-2217 using address fe80:0:0:0:201:29ff:fe8d:a2af%14.	GMT -5:00
12/19/23 16:48:20.900	Information	[Instrument Response] 0 Hello 1.0.20	GMT -5:00
12/19/23 16:48:20.978	Information	[Instrument Response] 0 Status 0 Connected to LICOR-CLX-2217	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Status 0 Cards: ddspcard motorcard	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Status 0 Mic_SN: 1809-0090	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 XRange 0 25	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 YRange 0 25	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 FocusRange 0 4	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Channels 700 800	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Intensity L2 L1.5 L1 L0.5 0.5 1 1.5 2 2.5 3 3.5 4 4.5 5 5.5 6 6.5 7 7.5 8 8.5 9 9.5 10	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Resolution 21um 42um 84um 169um 337um preview	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Quality lowest low medium high highest	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Response] 0 Camera Ready	GMT -5:00
12/19/23 16:48:21.010	Information	[Instrument Request]  8 ScanAutoEstimate 169um medium 0.0 0.1 5.0 0.0 4.9 700 800 	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Executing command: ScanAutoEstimate 169um medium 0 0.1 5 0 4.9 700 800	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Blocks=20	GMT -5:00
12/19/23 16:48:23.978	Information	[Instrument Response] 8 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=209	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 8 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 8 Result 0 OK	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic ADC1_Selftest                            2.05 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic ADC2_Selftest                            2.05 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Cooler_Voltage                    666 mV 	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_Temperature                       0.2 C  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Cooler_Voltage                    730 mV 	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           0 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_Temperature                       0.2 C  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_CPU_Box_Temperature                28 C  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Ambient_Scan_Box_Temperature               26 C  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Temperature                          25 C  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Microscope_Body_Temperature                32 C  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+3V_Digital_Supply                 3.31 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Digital_Supply                 5.00 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_ADC_Selftest                       2.05 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+10V_Analog_Supply               10.0 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Analog_Supply                5.06 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_+5V_Digital_Supply               4.93 V  	GMT -5:00
12/19/23 16:48:23.993	Information	[Instrument Response] 0 Status 0 Diagnostic Signals_-10V_Analog_Supply              -10.1 V  	GMT -5:00
12/19/23 16:48:24.040	Information	[Instrument Response] 0 Lid Closed	GMT -5:00
12/19/23 16:48:27.211	Information	Starting acquisition...	GMT -5:00
12/19/23 16:48:27.961	Information	[Instrument Request]  9 ScanAuto 169um medium 0.0 0.1 5.0 0.0 4.9 700 800 	GMT -5:00
12/19/23 16:48:28.008	Information	[Instrument Response] 9 Status 1 Executing command: ScanAuto 169um medium 0 0.1 5 0 4.9 700 800	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 XAlignments: global=-52 700=-0.61(0) 800=-7.22(-1)	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 X mapping: [x0=0.1,x1=5] => [left=-13,right=2315] w=290	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 YAlignments: global=42 700=-4.7(-1) 800=-3.17(-1)	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 Y mapping: [y0=0,y1=4.9] => [top=34,bottom=2354] h=290	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 Focus: mic=162, sys=-8, user=0 => 154	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 Computed scan parameters	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 Seconds/line=0.335775	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 y_scan_pixels=290, passes=2 => lines_to_scan=580	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 linesPerBlock=30 flushLinesPerBlock=15	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 Blocks=20	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Status 0 Estimates (sec): setup=2.02 line=0.355921 => total=209	GMT -5:00
12/19/23 16:48:28.024	Information	[Instrument Response] 9 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:48:28.039	Information	[Instrument Response] 9 Progress 0% Initializing	GMT -5:00
12/19/23 16:48:28.055	Information	[Instrument Response] 9 Status 0 Preparation complete	GMT -5:00
12/19/23 16:48:30.976	Information	[Instrument Response] 9 Status 0 Motors initialized	GMT -5:00
12/19/23 16:48:31.038	Information	[Instrument Response] 9 Status 0 X-Motor configured	GMT -5:00
12/19/23 16:48:31.038	Information	[Instrument Response] 9 Status 0 DSP configured	GMT -5:00
12/19/23 16:48:31.038	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        9.7 V  	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 Digital / DSP configured	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 Cooling	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 Pinged laser cooler disuse timer	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 Create MergeImageChannel: 700 20 0 0 290 15 290 290 xofs=1 yofs=0	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 Create MergeImageChannel: 800 20 0 0 290 15 290 290 xofs=0 yofs=0	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 start=34 scan_lines=15 line=0 end=290 700=Low,L1 800=High,4	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 start=34 scan_lines=30 line=0 end=290 700=High,4 800=Low,L1	GMT -5:00
12/19/23 16:48:31.085	Information	[Instrument Response] 9 Status 0 Move motors to initial positions	GMT -5:00
12/19/23 16:48:32.007	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:48:34.022	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:48:34.866	Information	[Instrument Response] 9 Progress 0% 4 minutes, 30 seconds	GMT -5:00
12/19/23 16:48:35.022	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         103 V  	GMT -5:00
12/19/23 16:48:35.084	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         161 V  	GMT -5:00
12/19/23 16:48:35.084	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:48:35.084	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:48:35.084	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:48:36.037	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:48:36.084	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:48:41.052	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/19/23 16:48:43.051	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:48:43.114	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         114 V  	GMT -5:00
12/19/23 16:48:43.114	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:48:43.114	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:48:44.067	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:48:50.190	Information	[Instrument Response] 9 ImageBlock 700 0 0 290 15 290 290 8700	GMT -5:00
12/19/23 16:48:50.190	Information	[Instrument Response] 9 ImageBlock 800 0 0 290 15 290 290 8700	GMT -5:00
12/19/23 16:48:50.190	Information	Transferring image block at (0,0) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:48:50.222	Information	Transferring image block at (0,0) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:48:55.923	Information	[Instrument Response] 9 Progress 8% 4 minutes	GMT -5:00
12/19/23 16:48:56.111	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         134 V  	GMT -5:00
12/19/23 16:48:56.173	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:48:56.173	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:48:56.173	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:48:56.173	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:48:57.111	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          91 V  	GMT -5:00
12/19/23 16:48:57.173	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:48:58.126	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:49:03.562	Information	[Instrument Response] 9 ImageBlock 700 0 15 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:03.562	Information	[Instrument Response] 9 ImageBlock 800 0 15 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:03.562	Information	Transferring image block at (0,15) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:49:03.594	Information	Transferring image block at (0,15) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:49:05.156	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      326 mA 	GMT -5:00
12/19/23 16:49:10.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:49:10.217	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:49:10.217	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:49:10.217	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:49:11.170	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:49:16.185	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:49:16.919	Information	[Instrument Response] 9 ImageBlock 700 0 30 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:16.919	Information	[Instrument Response] 9 ImageBlock 800 0 30 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:16.934	Information	Transferring image block at (0,30) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:49:16.950	Information	Transferring image block at (0,30) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:49:17.200	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:49:23.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         108 V  	GMT -5:00
12/19/23 16:49:23.292	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:49:23.292	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:49:23.292	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:49:24.230	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:49:26.916	Information	[Instrument Response] 9 Progress 19% 3 minutes, 30 seconds	GMT -5:00
12/19/23 16:49:30.291	Information	[Instrument Response] 9 ImageBlock 700 0 45 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:30.291	Information	[Instrument Response] 9 ImageBlock 800 0 45 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:30.291	Information	Transferring image block at (0,45) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:49:30.322	Information	Transferring image block at (0,45) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:49:36.289	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:49:36.352	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         127 V  	GMT -5:00
12/19/23 16:49:36.352	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:49:36.352	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:49:36.352	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:49:37.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:49:37.352	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:49:38.305	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:49:43.647	Information	[Instrument Response] 9 ImageBlock 700 0 60 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:43.647	Information	[Instrument Response] 9 ImageBlock 800 0 60 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:43.663	Information	Transferring image block at (0,60) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:49:43.678	Information	Transferring image block at (0,60) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:49:46.334	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:49:47.334	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:49:49.349	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:49:50.349	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          89 V  	GMT -5:00
12/19/23 16:49:50.411	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:49:50.411	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:49:50.411	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:49:50.411	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:49:51.364	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:49:52.364	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      295 mA 	GMT -5:00
12/19/23 16:49:56.675	Information	[Instrument Response] 9 Progress 31% 3 minutes	GMT -5:00
12/19/23 16:49:57.019	Information	[Instrument Response] 9 ImageBlock 700 0 75 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:57.019	Information	[Instrument Response] 9 ImageBlock 800 0 75 290 15 290 290 8700	GMT -5:00
12/19/23 16:49:57.019	Information	Transferring image block at (0,75) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:49:57.050	Information	Transferring image block at (0,75) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:50:03.408	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:50:03.471	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         104 V  	GMT -5:00
12/19/23 16:50:03.471	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:50:03.471	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:50:04.424	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:50:10.375	Information	[Instrument Response] 9 ImageBlock 700 0 90 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:10.375	Information	[Instrument Response] 9 ImageBlock 800 0 90 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:10.391	Information	Transferring image block at (0,90) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:50:10.407	Information	Transferring image block at (0,90) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:50:13.453	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:50:13.500	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:50:14.453	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:50:14.515	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:50:16.468	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         120 V  	GMT -5:00
12/19/23 16:50:16.515	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:50:16.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:50:16.515	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:50:17.468	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:50:18.467	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:50:19.483	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:50:21.482	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:50:22.482	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:50:23.747	Information	[Instrument Response] 9 ImageBlock 700 0 105 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:23.747	Information	[Instrument Response] 9 ImageBlock 800 0 105 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:23.747	Information	Transferring image block at (0,105) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:50:23.779	Information	Transferring image block at (0,105) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:50:26.825	Information	[Instrument Response] 9 Progress 43% 2 minutes, 30 seconds	GMT -5:00
12/19/23 16:50:29.527	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         151 V  	GMT -5:00
12/19/23 16:50:29.590	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         148 V  	GMT -5:00
12/19/23 16:50:29.590	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:50:29.590	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:50:29.590	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:50:30.527	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:50:30.590	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          86 V  	GMT -5:00
12/19/23 16:50:30.590	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:50:31.542	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:50:37.119	Information	[Instrument Response] 9 ImageBlock 700 0 120 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:37.119	Information	[Instrument Response] 9 ImageBlock 800 0 120 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:37.119	Information	Transferring image block at (0,120) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:50:37.135	Information	Transferring image block at (0,120) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:50:43.587	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          99 V  	GMT -5:00
12/19/23 16:50:43.633	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:50:43.633	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:50:43.633	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:50:44.586	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:50:45.586	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:50:46.602	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:50:50.476	Information	[Instrument Response] 9 ImageBlock 700 0 135 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:50.476	Information	[Instrument Response] 9 ImageBlock 800 0 135 290 15 290 290 8700	GMT -5:00
12/19/23 16:50:50.476	Information	Transferring image block at (0,135) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:50:50.507	Information	Transferring image block at (0,135) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:50:56.209	Information	[Instrument Response] 9 Progress 54% 2 minutes	GMT -5:00
12/19/23 16:50:56.646	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:50:56.693	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         117 V  	GMT -5:00
12/19/23 16:50:56.693	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:50:56.693	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:50:57.646	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:51:03.848	Information	[Instrument Response] 9 ImageBlock 700 0 150 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:03.848	Information	[Instrument Response] 9 ImageBlock 800 0 150 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:03.848	Information	Transferring image block at (0,150) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:51:03.863	Information	Transferring image block at (0,150) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:51:06.909	Information	[Instrument Response] 9 Progress 58% 1 minute, 50 seconds	GMT -5:00
12/19/23 16:51:09.690	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         140 V  	GMT -5:00
12/19/23 16:51:09.753	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:51:09.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:51:09.753	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:51:09.753	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:51:10.705	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          87 V  	GMT -5:00
12/19/23 16:51:10.752	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:51:11.705	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:51:15.720	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:51:15.767	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:51:16.720	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:51:16.782	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:51:16.876	Information	[Instrument Response] 9 Progress 62% 1 minute, 40 seconds	GMT -5:00
12/19/23 16:51:17.204	Information	[Instrument Response] 9 ImageBlock 700 0 165 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:17.204	Information	[Instrument Response] 9 ImageBlock 800 0 165 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:17.204	Information	Transferring image block at (0,165) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:51:17.235	Information	Transferring image block at (0,165) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:51:23.765	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:51:23.812	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:51:23.812	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:51:23.812	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:51:26.827	Information	[Instrument Response] 9 Progress 66% 1 minute, 30 seconds	GMT -5:00
12/19/23 16:51:30.576	Information	[Instrument Response] 9 ImageBlock 700 0 180 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:30.576	Information	[Instrument Response] 9 ImageBlock 800 0 180 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:30.576	Information	Transferring image block at (0,180) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:51:30.592	Information	Transferring image block at (0,180) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:51:36.293	Information	[Instrument Response] 9 Progress 70% 1 minute, 20 seconds	GMT -5:00
12/19/23 16:51:36.825	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         111 V  	GMT -5:00
12/19/23 16:51:36.887	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:51:36.887	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:51:36.887	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:51:37.824	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:51:43.932	Information	[Instrument Response] 9 ImageBlock 700 0 195 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:43.932	Information	[Instrument Response] 9 ImageBlock 800 0 195 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:43.948	Information	Transferring image block at (0,195) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:51:43.964	Information	Transferring image block at (0,195) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:51:45.869	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:51:45.916	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:51:46.635	Information	[Instrument Response] 9 Progress 74% 1 minute, 10 seconds	GMT -5:00
12/19/23 16:51:46.869	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:51:46.916	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:51:48.869	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:51:49.884	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:51:49.931	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         132 V  	GMT -5:00
12/19/23 16:51:49.931	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:51:49.931	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:51:49.931	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:51:49.931	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.00 V  	GMT -5:00
12/19/23 16:51:50.884	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          92 V  	GMT -5:00
12/19/23 16:51:50.931	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:51:50.931	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.93 V  	GMT -5:00
12/19/23 16:51:51.884	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:51:56.961	Information	[Instrument Response] 9 Progress 77% 1 minute	GMT -5:00
12/19/23 16:51:57.304	Information	[Instrument Response] 9 ImageBlock 700 0 210 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:57.304	Information	[Instrument Response] 9 ImageBlock 800 0 210 290 15 290 290 8700	GMT -5:00
12/19/23 16:51:57.304	Information	Transferring image block at (0,210) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:51:57.335	Information	Transferring image block at (0,210) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:52:03.943	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          92 V  	GMT -5:00
12/19/23 16:52:03.990	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:52:03.990	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:52:03.990	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:52:04.943	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:52:06.911	Information	[Instrument Response] 9 Progress 81% 50 seconds	GMT -5:00
12/19/23 16:52:10.661	Information	[Instrument Response] 9 ImageBlock 700 0 225 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:10.661	Information	[Instrument Response] 9 ImageBlock 800 0 225 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:10.676	Information	Transferring image block at (0,225) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:52:10.692	Information	Transferring image block at (0,225) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:52:12.973	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:52:13.988	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:52:15.988	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:52:16.050	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:52:16.394	Information	[Instrument Response] 9 Progress 85% 40 seconds	GMT -5:00
12/19/23 16:52:16.987	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:52:17.050	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         108 V  	GMT -5:00
12/19/23 16:52:17.050	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:52:17.050	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:52:17.050	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:52:17.050	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:52:18.003	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:52:21.002	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.1 V  	GMT -5:00
12/19/23 16:52:22.017	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:52:24.033	Information	[Instrument Response] 9 ImageBlock 700 0 240 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:24.033	Information	[Instrument Response] 9 ImageBlock 800 0 240 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:24.033	Information	Transferring image block at (0,240) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:52:24.064	Information	Transferring image block at (0,240) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:52:26.719	Information	[Instrument Response] 9 Progress 89% 30 seconds	GMT -5:00
12/19/23 16:52:30.062	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         125 V  	GMT -5:00
12/19/23 16:52:30.109	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:52:30.109	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          30 mA 	GMT -5:00
12/19/23 16:52:30.109	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:52:30.109	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:52:31.062	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          94 V  	GMT -5:00
12/19/23 16:52:31.109	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:52:36.686	Information	[Instrument Response] 9 Progress 93% 20 seconds	GMT -5:00
12/19/23 16:52:37.389	Information	[Instrument Response] 9 ImageBlock 700 0 255 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:37.405	Information	[Instrument Response] 9 ImageBlock 800 0 255 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:37.405	Information	Transferring image block at (0,255) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:52:37.420	Information	Transferring image block at (0,255) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:52:43.106	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.0 V  	GMT -5:00
12/19/23 16:52:44.106	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:52:44.169	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          89 V  	GMT -5:00
12/19/23 16:52:44.169	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:52:44.169	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:52:44.169	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:52:45.122	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:52:46.637	Information	[Instrument Response] 9 Progress 96% 10 seconds	GMT -5:00
12/19/23 16:52:48.121	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.2 V  	GMT -5:00
12/19/23 16:52:48.183	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  5.02 V  	GMT -5:00
12/19/23 16:52:49.136	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+12V_Analog_Supply                 12.3 V  	GMT -5:00
12/19/23 16:52:49.183	Information	[Instrument Response] 0 Status 0 Diagnostic Motor_+5V_Analog_Supply                  4.95 V  	GMT -5:00
12/19/23 16:52:50.761	Information	[Instrument Response] 9 ImageBlock 700 0 270 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:50.761	Information	[Instrument Response] 9 ImageBlock 800 0 270 290 15 290 290 8700	GMT -5:00
12/19/23 16:52:50.761	Information	Transferring image block at (0,270) with size 290x15 to the 700 channel.	GMT -5:00
12/19/23 16:52:50.792	Information	Transferring image block at (0,270) with size 290x15 to the 800 channel.	GMT -5:00
12/19/23 16:52:53.151	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         115 V  	GMT -5:00
12/19/23 16:52:53.213	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         155 V  	GMT -5:00
12/19/23 16:52:53.213	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          31 mA 	GMT -5:00
12/19/23 16:52:53.213	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          58 mA 	GMT -5:00
12/19/23 16:52:54.151	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          95 V  	GMT -5:00
12/19/23 16:52:56.588	Information	[Instrument Response] 9 Progress 100% 0 seconds	GMT -5:00
12/19/23 16:52:57.181	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         153 V  	GMT -5:00
12/19/23 16:52:57.228	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                         108 V  	GMT -5:00
12/19/23 16:52:57.228	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                          39 mA 	GMT -5:00
12/19/23 16:52:57.228	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                          34 mA 	GMT -5:00
12/19/23 16:52:58.181	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          96 V  	GMT -5:00
12/19/23 16:52:58.587	Information	[Instrument Response] 9 ImageBlock 700 0 285 290 5 290 290 2900	GMT -5:00
12/19/23 16:52:58.587	Information	[Instrument Response] 9 ImageBlock 800 0 285 290 5 290 290 2900	GMT -5:00
12/19/23 16:52:58.587	Information	Transferring image block at (0,285) with size 290x5 to the 700 channel.	GMT -5:00
12/19/23 16:52:58.587	Information	[Instrument Response] 9 Status 0 COMPLETE SCAN 263.713 seconds	GMT -5:00
12/19/23 16:52:58.634	Information	Transferring image block at (0,285) with size 290x5 to the 800 channel.	GMT -5:00
12/19/23 16:52:58.650	Information	[Instrument Response] 9 Status 0 Averages (sec): setup=2.00005 line=0.378814 => total=222	GMT -5:00
12/19/23 16:52:58.697	Information	[Instrument Response] 9 Status 0 Comparisons: setup=1.00997 line=0.939566	GMT -5:00
12/19/23 16:52:58.697	Information	[Instrument Response] 9 Status 0 Finalize motors	GMT -5:00
12/19/23 16:52:58.697	Information	[Instrument Response] 9 Progress 100% Scan complete	GMT -5:00
12/19/23 16:52:58.712	Information	[Instrument Response] 9 Result 0 OK	GMT -5:00
12/19/23 16:52:58.712	Information	Acquisition 0000034_01 finished with channels: 700 800.	GMT -5:00
12/19/23 16:52:58.947	Information	Setting the Camera Status panel back to the Ready state.	GMT -5:00
12/19/23 16:52:58.947	Information	Scan completed successfully	GMT -5:00
12/19/23 16:52:59.181	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                         109 V  	GMT -5:00
12/19/23 16:52:59.243	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          68 V  	GMT -5:00
12/19/23 16:52:59.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_700_Current                           1 mA 	GMT -5:00
12/19/23 16:52:59.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_800_Current                           1 mA 	GMT -5:00
12/19/23 16:52:59.243	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Power_Supply                        0.0 V  	GMT -5:00
12/19/23 16:53:00.196	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          66 V  	GMT -5:00
12/19/23 16:53:00.241	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          42 V  	GMT -5:00
12/19/23 16:53:01.191	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          41 V  	GMT -5:00
12/19/23 16:53:01.246	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          25 V  	GMT -5:00
12/19/23 16:53:02.201	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          26 V  	GMT -5:00
12/19/23 16:53:02.261	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          17 V  	GMT -5:00
12/19/23 16:53:03.193	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          18 V  	GMT -5:00
12/19/23 16:53:03.262	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                          11 V  	GMT -5:00
12/19/23 16:53:04.231	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                          12 V  	GMT -5:00
12/19/23 16:53:04.271	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           8 V  	GMT -5:00
12/19/23 16:53:05.218	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           9 V  	GMT -5:00
12/19/23 16:53:05.287	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           6 V  	GMT -5:00
12/19/23 16:53:06.205	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           6 V  	GMT -5:00
12/19/23 16:53:06.274	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           3 V  	GMT -5:00
12/19/23 16:53:07.224	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           3 V  	GMT -5:00
12/19/23 16:53:07.277	Information	[Instrument Response] 0 Status 0 Diagnostic Laser_Cooler_Current                      265 mA 	GMT -5:00
12/19/23 16:53:08.227	Information	[Instrument Response] 0 Status 0 Diagnostic APD_800_HV_Supply                           1 V  	GMT -5:00
12/19/23 16:53:10.232	Information	[Instrument Response] 0 Status 0 Diagnostic APD_700_HV_Supply                           1 V  	GMT -5:00
12/19/23 16:55:12.617	Information	[Instrument Request]  10 Stop 	GMT -5:00
12/19/23 16:55:12.617	Information	[Instrument Response] 10 Result 0 OK	GMT -5:00
12/19/23 16:55:13.710	Information	[Instrument Response] 0 Status 1 Closing connection	GMT -5:00
